GE Healthcare™ Recalls Giraffe infant warmers and Panda i-Res infant warmers
GE Healthcare, LLC is announcing a recall on their Giraffe and Panda i-Res infant warmers due to bedside panels and latch areas cracking or breaking.
The Giraffe and Panda i-Res infant warmers are currently under recall as of July 12th, 2019 because the bedside panels and latch areas of the warmers can crack or break if the unit is moved using the bedside panels instead of the front or rear maneuvering handle. This is extremely important – if an infant comes into contact with a bedside panel with a cracked or broken latch, the panel can disengage and fall open, no longer protecting the infant from falling off of the warmer.
The following products are affected by this recall as the latch and bedside panels have begun cracking or breaking:
All GE Healthcare infant warmer units from US Med-Equip (USME) are currently being inspected and labeled for any latch cracking or breaking before they go on rent. The units that are currently being rented and coming off of rent are also being inspected and labeled for any damages pertaining to this recall. If you have an affected infant warmer from USME, please email email@example.com to place your report and get your unit replaced.
GE Healthcare’s Maintenance Services
Phone hours: 7:00 am to 5:00 pm PT, Monday – Friday
For any affected infant warmers not coming from USME, complete and return the Customer Response form
by e-mail to MIC.Recall@ge.com indicating what components require replacement.
GE Healthcare Products & Services
Phone hours: 7:00am to 5:00pm PT Monday – Friday
Our Mission is to promote the healing process by providing smart solutions for our partners and the patients they serve.
We are committed to your patients as much as you are — it’s rooted in our daily actions — so we understand the importance of safely servicing recalls promptly.
For any further questions regarding equipment recalls, please reach out to firstname.lastname@example.org.